For a simple like-for-like replacement, you should perform installation qualifications (IQ) and operational qualifications (OQ) on the new equipment.
The next step in the contract testing labs process would be to execute a performance qualification (PQ) and a comparison protocol. Any pharma contract manufacturing teams should be able to demonstrate the results obtained with new equipment are the same from both an efficacy and quality perspective.
In many cases, the PQ and comparison may be combined into one document. If the results from the protocol match up, you can decommission the old contract lab testing equipment and start using the new equipment.
When changing from one contract laboratory technology to a different, more sensitive technology, there are more decisions to be considered. You need to work closely with your client on the lab testing specifics of their product so you can perform the transition with minimal disruption to their and your product flow.
These more complex upgrades require a lot of upfront planning. As with the like-for-like replacement, you need to make sure the current contract laboratory equipment is operating appropriately.
As before, you need to perform the IQ and PQ on the new equipment. It is in the PQ and comparison protocol where attention to detail will be crucial to execution. For instance, when changing from high-performance liquid chromatography (HPLC) to the more sensitive ultra-high pressure liquid chromatography (UHPLC), you need to be prepared to address the possibility that you could see some unknown peaks during the analysis.
These contract testing peaks need to be identified and characterized before you can continue with the transfer. While this investigation is underway, you need to determine the suitability of the contract pharma product for release. This includes understanding the potential impact the peaks might have on stability of the item being analyzed.
If the method being converted is a US Pharmacopeial Convention (USP) method, you need to be sure that the new methodology generates results that are equivalent or better than the results obtained using the compendial method. The comparability protocol details will help you address the unforeseen circumstances should they occur during the execution of the protocol.
The issues arising from the conversion of one method to a more sensitive method need to be fully addressed and explained before decommissioning the obsolete equipment.
There may be situations where you will need to maintain both the new and the old lab testing equipment operational at the same time. For instance, you may want to continue to monitor the stability of a product on the equipment the analysis was started on and not change to the new equipment until the last stability time point has been analyzed.
In this case, you will want to maintain the old equipment while utilizing the new equipment for release and future stability monitoring. These situations need to be discussed with your client and included in the laboratory timeline for transitioning the laboratory.
Another important element of transitioning a contract test laboratory is making sure your analysts are properly trained on the new equipment while maintaining expertise on the old equipment until the transition is complete. Many companies selling analytical equipment have experts to perform the necessary training and to help address issues that may come up as the analysts become more familiar with the equipment.
It is important for the laboratory and your clients that analysts are properly trained so they can deal more effectively with problems that occur during the transition period. The more training and familiarity the analysts have in running the new equipment, the less disruption there will be to operations.
When introducing new contract laboratory equipment and new methodologies, it is important to have a comprehensive implementation plan that discusses what equipment you will introduce. The lab testing timeline for the introduction of the equipment and how you will investigate unexpected analytical results are important during the transition. Additionally, how you will make sure your analysts are appropriately trained is essential to cause minimal interruptions to operations.
Putting your contract laboratory plan down in writing and discussing it with your clients up front will save you a lot of time and minimize delays as you implement your plan.
Vol. 41, No. 6
Regulatory Compliance Associates® (RCA) provides worldwide services to the following industries for resolution of compliance and regulatory challenges:
We understand the complexities of running a life science business and possess areas of expertise that include every facet of R&D, operations, regulatory affairs, quality, and manufacturing. We are used to working on the front lines and thriving in the scrutiny of FDA-and globally-regulated companies. As your partners, we can negotiate the potential minefield of regulatory compliance and private equity due diligence with insight, hindsight, and the clear advantage of our unique expertise and experience.
- Founded in 2000
- Headquartered in Wisconsin (USA)
- Regional offices in Florida, Colorado and Europe
- Expertise backed by over 500 industry subject matter experts
- Acquired by Sotera Health in 2021